HeartStart AEDs are designed for the ordinary person in the extraordinary moment. It is easy to set-up and includes automatic Life Guidance features like voice prompts and CPR coaching to help guide you through the treatment of sudden cardiac arrest.
Ready to use when you need it
Setting up the AED is easy and quick. The pads and battery are already installed. Just pull the green tab to launch the initial self-test.
Fast shocks to improve treatment success
Since minimizing the time to shock after the last CPR compression may improve survival, shock delivery time is critical.
Personalized therapy for each patient
Before you can deliver a shock, the AED automatically checks to make sure a shock is necessary and that the right therapy is given. If you install the infant/child SMART Pads, the defibrillator will automatically adjust to a lower energy level and modify the coaching instructions to provide infant/child CPR.
9 volt DC, 4.2 Ah, composed of disposable long-life lithium manganese dioxide primary cells
Minimum 200 shocks or 4 hours of operating time (EN60601-2-4:2003)
Battery is labeled with an install-by date of at least five years from date of manufacture
Four years typical when battery is installed by the install-by date. (Will power the AED in standby state within the specified Standby Temperature range, assuming one battery insertion test and no defibrillation uses)
Defibrillator, battery (one, pre-installed), SMART Pads (one set, pre-installed), set-up/maintenance guide with expiration date tags, owner's manual, quick reference guide
SMART Truncated exponential biphasic. Waveform parameters adjusted as a function of patient impedance
Defibrillation peak current - Adult
32 A (150 J nominal) into a 50 ohm load
Defibrillation peak current - Pediatric
19 A (50 J nominal) into a 50 ohm load (using optional infant/child SMART Pads Cartridge)
Shock-to-Shock Cycle Time
Typically less than 20 seconds between shocks in a series
Able to deliver a shock after the end of a CPR interval, typically in eight seconds
Detailed voice messages guide responder through use of the defibrillator
Via adhesive pads placed on patient's bare skin as illustrated on pads
Green SMART Pads cartridge handle, green On/Off button, blue i-button, orange Shock button
Ready light, blue i-button, caution light
NOTE: Transportation charges are not included in the item price or charges. These charges are the buyer’s responsibility.
CLICK: Download and Read Full Digital Edition of your Daily Newspaper on your Mobile Device
NEW: Advanced Cardiovascular Life Support Course (ACLS) 2018 Training Schedule
CLICK: American Heart Association AHA Basic Life Support BLS Course 2018 Schedule
2018 TRAINING SCHEDULE: American Heart Association First Aid CPR AED Course
Join the Medicalworld Nigeria Telegram group for latest Medical updates, news and jobs!!
NOW YOU CAN BE THE REPORTER: Submit your story, news reports, articles, pictures, events and press release to us via Watsapp/ Telegram (08105233399) or Email (email@example.com) | Follow us on Twitter @Medicalworld
Medicalworld Nigeria Recruiting Medical Professionals for an Oil and Gas Contracting Firm ITT. Click HERE to Apply!!